The aerosol impact spectrometer: a versatile platform for studying the velocity dependence of nanoparticle-surface impact phenomena
© The Author(s); licensee Springer on behalf of EPJ. 2017
Received: 12 December 2016
Accepted: 9 March 2017
Published: 11 April 2017
A new apparatus designed to accelerate/decelerate and study the surface impact phenomena of charged aerosols and nanoparticles over a wide range of mass-to-charge (m/z) ratios and final velocities is described. A nanoparticle ion source coupled with a linear electrostatic trap configured as an image charge detection (ICD) mass spectrometer allows determination of the mass-to-charge ratio and the absolute charge and mass of single nanoparticles. A nine-stage linear accelerator/decelerator is used to fix the final velocity of the nanoparticles, and in the results reported here the coefficient of restitution for polystyrene latex spheres (PSLs) impacting on silicon is measured using ICD techniques. To enable this apparatus to study a wide range of m/z, the data acquisition system uses a transient digitizer interfaced to a field-programmable gate array module that allows real time calculation of m/z and determination of the pulse sequence for the linear accelerator/decelerator. Electrospray ionization of a colloidal suspension of PSL spheres of 510 and 990 nm has been used to demonstrate acceleration and deceleration of charged nanoparticles and the resolution of the apparatus. Measurements of the coefficient of restitution for PSLs on silicon over the range 10-400 m/s are consistent with previous studies.
KeywordsImage charge detection Nanoparticle acceleration Deceleration Coefficient of restitution
As interest in the characterization of nanoparticles, aerosols and dusts increases, the need for the development of new tools for the manipulation and analysis of single particles continues to grow . An area of continued interest for an understanding of problems ranging from atmospheric chemistry to astrophysical phenomena to industrial applications pertains to the impact phenomena of nanoparticles. Theoretical and experimental studies of nanoparticle-surface collisions at both low and high velocity ranges continue to be reported [2–7]. There is a large body of work on hypervelocity impact phenomena dating back to the early 1960’s motivated by the need to understand the effect of cosmic dust and micrometeoroid impacts on space vehicles. Larger objects are most conveniently accelerated with light-gas guns , however, for smaller particles that can be easily charged, electrostatic accelerators are a convenient approach [9–12]. In early work in this area, a 2 MV van de Graaff dust accelerator was described and shown to accelerate 1 μm particles to ~ 6 km/s [10, 11]. One early example of a switched, multistage linear accelerator for large particles was described by Vedder , however it was not until advances in high-voltage switching circuitry that laboratory-scale linear accelerators, such as the one described by Hendell and Even , became more common. In the present work the development of a versatile new nanoparticle mass spectrometer/accelerator/decelerator, the Aerosol Impact Spectrometer (AIS), is described with a demonstration of its capabilities using polystyrene latex (PSL) spheres, including studies of collision inelasticity on silicon wafers by determination of incident and scattered velocities in measurements of the coefficient of restitution.
The key to carrying out mass spectrometric measurements on single charged nanoparticles is the use of charge detection mass spectrometry (CDMS) techniques (1). CDMS determines the absolute charge on a particle from the magnitude of the image charge induced on a pickup electrode when a charged particle passes through. The image charge waveform also yields the particle time-of-flight (TOF) and velocity through the pickup providing the mass-to-charge ratio for fixed energy particles . This method has been used in accelerator experiments since at least 1960 , and with the increasing interest in studies of massive biomolecules, cells and nanoparticles it has seen more recent applications in mass spectrometry measurements [16–24], with the state of the art recently described by Keifer and Jarrold . In our own laboratory, we make use of a variant of this approach, applying a charge-pickup electrode in an electrostatic fast-ion beam trap to monitor the ion density oscillating in the trap and carry out Fourier-transform (FT) mass spectrometry on ensembles of molecular ions [26, 27]. In the present apparatus the m/z ratio of a single particle provides the information required to accelerate or decelerate that particle for studies of particle impact.
As shown by Hendell and Even , and later applied by Hsu and co-workers , a linear accelerator for large molecular ions can be configured using modern high-voltage MOSFET switching techniques. In the present apparatus, the CDMS determines the m/z ratio for each particle in real time, providing the information required to accelerate or decelerate single particles over a wide range of m/z by generating the appropriate acceleration/deceleration pulsed waveform for a linear accelerator (LINAC) structure composed of a series of cylindrical electrostatic elements. Acceleration of ion ensembles by this technique is not hindered by space charge limitations since the effect of space charge is progressively reduced as the particles accelerate. Deceleration of ion ensembles does become problematic as a result of increasing space charge, and that is the benefit of working with single charged nanoparticles where this limitation is not present.
In the following sections, the initial results obtained with the AIS will be presented, showing the measurement of charge/mass distributions, acceleration and deceleration of single charged PSL spheres, and measurements of impact inelasticity of single particles with a silicon substrate. These results show that the AIS will have great utility for measuring the reflection of nanoparticles from surfaces, yielding quantitative information on the coefficient of restitution that describes the inelasticity of such collisions [29, 30].
Charged particles are formed by electrospray ionization and transferred into a low vacuum through a 150-μm Pt electron microscope aperture. A heater tube dries the particles before they enter the ADL, which collimates and focuses the particle beam. The particle beam passes through two stages of differential pumping and enters a chamber that houses the electrostatic quadrupole deflector. The QD selectively turns particles 90° based on their kinetic energy per charge into the next chamber that contains the NET. The NET is a linear electrostatic trap configured as an image charge mass spectrometer, as first described by Zajfman and co-workers on ion ensembles [31, 32] and in a single-particle application by Benner . Briefly, the NET is gated to trap one particle at a time, before measuring the mass-to-charge ratio and the absolute charge of the particle. These particles can be trapped in the NET with an efficiency ~70% for periods in excess of 5 seconds with a distribution of oscillation frequencies in the several hundred Hertz range. A Labview-based data acquisition code uses the m/z data to calculate the required switching times for the LINAC to accelerate the particle up to the required velocity. The particle is then released from the trap into the LINAC for acceleration to the desired velocity.
Electrospray ionization (ESI) source
The use of ESI for this type of application is based on prior work by Benner, Austin and others on polystyrene latex spheres, silica, mineral dusts and other systems [33–36]. The ESI source consists of a 360 μm O.D., 75 μm I.D. fused silica capillary mounted on an isolated aluminum block, which is held at approximately +4–6 kV. To date the AIS instrument has only been run in positive mode, but no impediment is anticipated for using it to study negatively charged particles. The tip of the capillary is positioned ~10 mm in front of a 150 μm aperture and is enclosed within a glass tube. The aperture is mounted on a 6.4 mm OD, 3.86 mm ID stainless steel tube that passes through an Ultra-Torr fitting into the instrument. The portion of this tube that extends out of the vacuum chamber is surrounded by an enclosure, through which nitrogen, heated to ~85 °C, flows. The heated nitrogen helps to desolvate the particles, and also acts as a curtain gas. 510 nm (Polyscience #07307) and 990 nm (Polyscience #07310) PSL spheres were suspended in 1:1 mixtures of 25 mM aqueous ammonium acetate and methanol, then delivered to the electrospray tip by a syringe pump operating at a rate of 0.26 mL/h. The final number densities were 30 × 109 particles per mL for the 510 nm PSL suspension, and 4.1 × 108 particles per mL for the 990 nm suspension. After entering the vacuum chamber, the particles pass through a 120 mm long, 4.5 mm I.D. stainless steel tube, which is heated to ~185 °C to completely desolvate the particles, and to evaporate charged solvent droplets.
Aerodynamic lens (ADL)
After exiting the heater tube, the particles enter an ADL, which aerodynamically focuses and collimates the beam of particles [37, 38]. In this instance the ADL consists of 5 apertures separated by 59 mm long, 12.1 mm I.D. spacers. The diameters of the apertures are 5.9, 5.3, 4.6, 3.4 and 3.1 mm. The section after the 5th aperture is connected to a mechanical vacuum pump. The pressure at the start of the ADL is 2.6 Torr, and the pressure in the pumped region is typically in the range of 0.5-1 Torr. The precise pressure after the ADL is adjusted, by throttling the vacuum pump, in order to maximize transmission of particles in a specific size range. Calculations along the line of those described by Wang and McMurry indicate that this ADL should transmit over 95% of particles in the range of 75 – 1200 nm.
Quadrupole deflector (QD)
Following the ADL, the particles pass through a 3.1 mm diameter aperture into the first differential pumping stage, which is connected to a Roots blower backed by a rotary mechanical pump. This pumping stage also contains an image charge detector tube (ICD) to confirm transmission of particles through the aerodynamic lens. A 5 mm aperture leads to the second differential pumping stage, pumped by a Pfeiffer TMH 064 turbomolecular pump. A 3.2 mm aperture separates the second differential pumping stage from the quadrupole deflector chamber, which is pumped by an Osaka TG240 turbomolecular pump. The quadrupole deflector chamber is pumped to a vacuum of 3 × 10-5 Torr.
The QD assembly is in the center of the chamber and consists of 4 parallel, quarter-cylinder stainless steel rods (19 mm radius). The rods are mounted on 1/8″ precision ground glass spheres that locate the rods such that their curved faces are tangential to an inscribed circle of 33.7 mm diameter. Positive and negative potentials are applied to opposing pairs of rods, resulting in an electrostatic field that will turn particles that have kinetic energy, in eV per charge, equivalent to the potential applied to the rods. Particles with excess kinetic energy will overshoot the bend, whereas particles with less kinetic energy will impact the electrode on the inside of the curve.
There are two sets of ion optics in the chamber with the QD, each comprised of an einzel lens, an x-y deflector and an ICD, modeled after the detector presented by Fuerstenau and Benner . One is positioned before the entrance to the QD, the other after the exit from the QD. These ion optics serve to focus the selected particles and direct them to the next chamber, which contains the NET and LINAC. The first ICD (ICD-QD1) is used to confirm transmission of particles through the ADL. The second ICD (ICD-QD2) is used to determine the optimal deflector potential to direct the particle beam towards the next chamber.
Nanoparticle electrostatic trap
This data is measured and calculated on the fly for each particle and is used to create the timing sequence used by the HV switches for acceleration/deceleration.
The first three electrodes act as a lens with each element connected to external power supplies (two KIKUSUI PMC350-0.2A and one Canberra 3002). The remaining 9 electrodes are wired through two HV feedthroughs (30 kV) with every second element collectively wired to one feedthrough and every other element collectively wired to the other feedthrough. All elements in each set are connected together with copper rods. These two rods are connected to each HV feedthrough with a shielded HV cable. The elements are pulsed with two 30 kV HV switches (Behlke 301-03-GSM).
In Eq (4), t x is the time at position x, v x is the velocity at position x, E x is the electric field at position x, and dx is the step size of the calculation, in this case 0.01 mm. This timing calculation is performed while the particle is still trapped in the NET. Upon completion of the calculation, timing data is transferred to the FPGA. The FPGA then releases the particle from the trap by lowering the exit mirror and triggers the two high voltage (HV) switches connected to the LINAC at the times required to accelerate/decelerate the particle to a final energy that depends on the number of elements used and the potential applied to each element.
After acceleration, the FPGA digitizer captures a waveform from the output of ICD3. The waveform is transferred to the Labview program which calculates the accelerated velocity of the particle. The program also determines if the particle has rebounded from the collision target, and calculates the rebound velocity of the particle from the rebounding peak width. After completing this final acquisition the program saves all information to a data file and resets itself and the FPGA to accept a new particle. The entire run time for each particle in this experiment was approximately ~200 ms.
Results and discussion
Charge distributions in 510 and 990 nm PSL colloids
Variable acceleration and deceleration of charged nanoparticles
Impact dynamics and coefficient of restitution for PSL nanoparticles
Formation, trapping, acceleration/deceleration, and coefficient of restitution measurements of highly-charged submicron particles have been demonstrated. Single highly charged PSL spheres have been generated with an ESI source. Subsequent trapping and CDMS analysis of individual particles has been demonstrated. Individual particle acceleration/deceleration has been demonstrated, allowing the acquisition of quantitative coefficient of restitution data. The flexibility of single particle on-the-fly analysis allows the Aerosol Impact Spectrometer to function with a wide range of nanoparticle masses and charges from a given particle source. Additionally, the variable energy selection of the spectrometer allows a variety of particle sources to be implemented in addition to the demonstrated electrospray ionization, including liquid metal ion sources[54–56] and needle-charge dust sources [57, 58]. The wide range of final energies achievable with the variable linear accelerator/decelerator will allow for a variety of scattering experiments to be performed to examine both hypo- and hypervelocity impact phenomena
- e :
Coefficient of restitution
Field programmable gate array
Image charge detector
Nanoparticle electrostatic trap
This work was supported by the NSF Major Research Instrumentation Instrument Development Program under grant number CHE-1229690 and the Center for Aerosol Impacts on Climate and the Environment (CAICE), a National Science Foundation Center for Chemical Innovation under grant number CHE-1305427. We acknowledge contributions to this project by R. Otto, J.J. Rivera, K.A. Nadler, C.K. Anderson and J. Taulane, as well as discussions with S. De Dea and partial support from Cymer, an ASML Company.
The authors declare that they have no competing interests.
BA and MM participated in the design of the apparatus including the data acquisition system, carried out the measurements and drafted the manuscript. RC conceived of the study, participated in design and coordination and helped draft the manuscript. All authors read and approved the final manuscript.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.
- Benner WH (1997) A gated electrostatic ion trap to repetitiously measure the charge and m/z of large electrospray ions. Anal Chem 69:4162–4168View ArticleGoogle Scholar
- Xie J, Dong M, Li S (2016) Dynamic impact model of plastic deformation between micro-particles and flat surfaces without adhesion. Aerosol Sci Technol 50:321–330View ArticleGoogle Scholar
- Dunn PF, Brach RM, Caylor MJ (1995) Experiments on the low-velocity impact of microspheres with planar surfaces. Aerosol Sci Technol 23:80–95View ArticleGoogle Scholar
- Burchell MJ, Armes SP (2011) Impact ionisation spectra from hypervelocity impacts using aliphatic poly(methyl methacrylate) microparticle projectiles. Rapid Commun Mass Spectrom 25:543–550View ArticleGoogle Scholar
- Mocker A, Bugiel S, Auer S, Baust G, Colette A, Drake K et al (2011) A 2 MV Van de Graaff accelerator as a tool for planetary and impact physics research. Rev Sci Instrum 82:095111ADSView ArticleGoogle Scholar
- Kuuluvainen H, Arffman A, Saukko E, Virtanen A, Keskinen J (2013) A new method for characterizing the bounce and charge transfer properties of nanoparticles. J Aerosol Sci 55:104–115View ArticleGoogle Scholar
- Visser CW, Frommhold PE, Wildeman S, Mettin R, Lohse D, Sun C (2015) Dynamics of high-speed micro-drop impact: numerical simulations and experiments at frame-to-frame times below 100 ns. Soft Matter 11:1708–1722ADSView ArticleGoogle Scholar
- Motoyashiki Y, Hasegawa S, Okudaira K, Sato E (2008) Micrometeoroid impact on ceramic thin components for interplanetary probe. Int J Impact Eng 35:1666–1671View ArticleGoogle Scholar
- Shelton H, Hendricks CD, Wuerker RF (1960) Electrostatic acceleration of microparticles to hypervelocities. J Appl Phys 31:1243–1246ADSView ArticleGoogle Scholar
- Friichtenicht JF (1962) Two-million-volt electrostatic accelerator for hypervelocity research. Rev Sci Instrum 33:209–212ADSView ArticleGoogle Scholar
- Friichtenicht JF (1964) Micrometeoroid simulation using nuclear accelerator techniques. Nucl Inst Methods 28:70–78ADSView ArticleGoogle Scholar
- Keaton PW, Idzorek GC, Rowton LJ, Seagrave JD, Stradling GL, Bergeson SD et al (1990) A hypervelocity - microparticle - impacts laboratory with 100 km/s projectiles. Int J Impact Eng 10:295–308View ArticleGoogle Scholar
- Vedder JF (1978) Microparticle accelerator of unique design. Rev Sci Instrum 49:1–7ADSView ArticleGoogle Scholar
- Hendell E, Even U (1995) Tabletop linear accelerator for massive molecules. Rev Sci Instrum 66:3901–3902ADSView ArticleGoogle Scholar
- Gamero-Castaño M (2007) Induction charge detector with multiple sensing stages. Rev Sci Instrum 78:043301ADSView ArticleGoogle Scholar
- Fuerstenau SD, Benner WH (1995) Molecular weight determination of megadalton DNA electrospray ions using charge detection time-of-flight mass spectrometry. Rapid Commun Mass Spectrom 9:1528–1538View ArticleGoogle Scholar
- Schultz JC, Hack CA, Benner WH (1998) Mass determination of megadalton-DNA electrospray ions using charge detection mass spectrometry. J Am Soc Mass Spectrom 9:305–313View ArticleGoogle Scholar
- Smith JW, Siegel EE, Maze JT, Jarrold MF (2011) Image charge detection mass spectrometry: Pushing the envelope with sensitivity and accuracy. Anal Chem 83:950–956View ArticleGoogle Scholar
- Doussineau T, Antoine R, Santacreu M, Dugourd P (2012) Pushing the limit of infrared multiphoton dissociation to megadalton-size DNA ions. J Phys Chem Lett 3:2141–2145View ArticleGoogle Scholar
- Contino NC, Pierson EE, Keifer DZ, Jarrold MF (2013) Charge detection mass spectrometry with resolved charge states. J Am Soc Mass Spectrom 24:101–108View ArticleGoogle Scholar
- Keifer DZ, Shinholt DL, Jarrold MF (2015) Charge detection mass spectrometry with almost perfect charge accuracy. Anal Chem 87:10330–10337View ArticleGoogle Scholar
- Doussineau T, Mathevon C, Altamura L, Vendrely C, Dugourd P, Forge V et al (2016) Mass determination of entire amyloid fibrils by using mass spectrometry. Angew Chem Int Ed 55:2340–2344View ArticleGoogle Scholar
- Barney BL, Pratt SN, Austin DE (2016) Survivability of bare, individual Bacillus subtilis spores to high-velocity surface impact: Implications for microbial transfer through space. Planet Space Sci 125:20–26ADSView ArticleGoogle Scholar
- Elliott AG, Merenbloom SI, Chakrabarty S, Williams ER. (2017) Single particle analyzer of mass: A charge detection massspectrometer with a multi-detector electrostatic ion trap. Int J Mass Spec. 414:44-55
- Keifer DZ, Jarrold MF. Single-molecule mass spectrometry. (2016) Mass Spec Rev. http://dx.doi.org/10.1002/mas.21495
- Ring S, Pedersen HB, Heber O, Rappaport ML, Witte PD, Bhushan KG et al (2000) Fourier transform time-of-flight mass spectrometry in an electrostatic ion beam trap. Anal Chem 72:4041–4046View ArticleGoogle Scholar
- Johnson CJ, Shen BB, Poad BLJ, Continetti RE (2011) Photoelectron-photofragment coincidence spectroscopy in a cryogenically cooled electrostatic ion beam trap. Rev Sci Instrum 82:105105ADSView ArticleGoogle Scholar
- Hsu YF, Lin JL, Lai SH, Chu ML, Wang YS, Chen CH (2012) Macromolecular Ion Accelerator. Anal Chem 84:5765–5769View ArticleGoogle Scholar
- Schoner C, Rennecke S, Weber AP, Poschl T (2014) Introduction of a new technique to measure the coefficient of restitution for nanoparticles. Chem Ing Tech 86:365–374View ArticleGoogle Scholar
- Eurov DA, Kurdyukov DA, Stovpiaga EY, Salasyuk AS, Jager J, Scherbakov AV et al (2014) Hypersonic properties of monodisperse spherical mesoporous silica particles. J Phys D Appl Phys 47:335303View ArticleGoogle Scholar
- Zajfman D, Heber O, Vejby-Christensen L, Ben-Itzhak I, Rappaport M, Fishman R et al (1997) Electrostatic bottle for long-time storage of fast ion beams. Phys Rev A 55:R1577–R1580ADSView ArticleGoogle Scholar
- Dahan M, Fishman R, Heber O, Rappaport M, Altstein N, Zajfman D et al (1998) A new type of electrostatic ion trap for storage of fast ion beams. Rev Sci Instrum 69:76–83ADSView ArticleGoogle Scholar
- Bogan MJ, Benner WH, Hau-Riege SP, Chapman HN, Frank M (2007) Aerosol sample preparation methods for X-ray diffractive imaging: Size-selected spherical nanoparticles on silicon nitride foils. Aerosol Sci 38:1119–1128View ArticleGoogle Scholar
- Benner WH, Bogan MJ, Rohner U, Boutet S, Woods B, Frank M (2008) Non-destructive characterization and alignment of aerodynamically focused particle beams using single particle detection. Aerosol Sci 39:917–928View ArticleGoogle Scholar
- Kerby JD, Terik Daly R, Austin DE (2013) A novel particle source based on electrospray charging for dust accelerators and its significance for cosmic dust studies. Earth Planets Space 65:157–165ADSView ArticleGoogle Scholar
- Daly RT, Kerby JD, Austin DE (2013) Electrospray charging of minerals and ices for hypervelocity impact research. Planet Space Sci 75:182–187ADSView ArticleGoogle Scholar
- Liu P, Ziemann PJ, Kittelson DB (1995) Generating particle beams of controlled dimensions and divergence: I. Theory of particle motion in aerodynamic lenses and nozzle expansions. Aerosp Sci Technol 22:293–313View ArticleGoogle Scholar
- Liu P, Ziemann PJ, Kittelson DB (1995) Generating particle beams of controlled dimensions and divergence: II. Experimental evaluation of particle motion in aerodynamic lenses and nozzle expansions. Aerosp Sci Technol 22:314–324View ArticleGoogle Scholar
- Wang X, McMurry PH (2006) A design tool for aerodynamic lens systems. Aerosp Sci Technol 40:320–334View ArticleGoogle Scholar
- Dahl DA (2000) SIMION for the personal computer in reflection. Int J Mass Spectrom 200:3–25View ArticleGoogle Scholar
- Cai Y, Peng WP, Kuo SJ, Lee YT, Chang HC (2002) Single-Particle Mass Spectrometry of Polystyrene Microspheres and Diamond Nanocrystals. Anal Chem 74:232–238View ArticleGoogle Scholar
- Doussineau T, Santacreu M, Antoine R, Dugourd P, Zhang W, Chaduc I et al (2013) The charging of micellar nanoparticles in electrospray ionization. ChemPhysChem 14:603–609View ArticleGoogle Scholar
- Ouadah N, Doussineau T, Hamada T, Dugourd P, Bordes C, Antoine R (2013) Correlation between the charge of polymer particles in solution and in the gas phase investigated by zeta-potential measurements and electrospray ionization mass spectrometry. Langmuir 29:14074–14081View ArticleGoogle Scholar
- Rayleigh LXX (1882) On the equilibrium of liquid conducting masses charged with electricity. Philos Mag Ser 5 14:184–186View ArticleGoogle Scholar
- Fenn JB, Mann M, Kai Meng C, Fu Wong S, Whitehouse CM (1989) Electrospray ionization for mass spectrometry of large biomolecules. Sci New Ser 246:64–71Google Scholar
- Vazquez G, Alvarez E, Navaza JM (1995) Surface tension of alcohol water + water from 20–50 degrees C. J Chem Eng Data 40:611–614View ArticleGoogle Scholar
- Doussineau T, Desert A, Lambert O, Taveau JC, Lansalot M, Dugourd P et al (2015) Charge detection mass spectrometry for the characterization of mass and surface area of composite nanoparticles. J Phys Chem C 119:10844–10849View ArticleGoogle Scholar
- Wall S, John W, Wang H-C, Goren SL (1990) Measurements of kinetic energy loss for particles impacting surfaces. Aerosol Sci Technol 12:926–946View ArticleGoogle Scholar
- Tsai C-J, Pui DYH, Liu BYH (1990) Capture and rebound of small particles upon impact with solid surfaces. Aerosol Sci Technol 12:497–507View ArticleGoogle Scholar
- Dahneke B (1995) Particle bounce or capture - search for an adequate theory: I. Conservation-energy-model for a simple collision process. Aerosol Sci Technol 23:25–39View ArticleGoogle Scholar
- Andres RP (1995) Inelastic energy transfer in particle/surface collisions. Aerosol Sci Technol 23:40–50View ArticleGoogle Scholar
- Valentini P, Dumitrica T (2007) Microscopic theory for nanoparticle-surface collisions in crystalline silicon. Phys Rev B 75:224106ADSView ArticleGoogle Scholar
- Dahneke B (1975) Further measurements of the bouncing of small latex spheres. J Colloid Interface Sci 51:58–65View ArticleGoogle Scholar
- Vladimirov VV, Badan VE, Gorshkov VN, Soloshenko IA (1993) Liquid metal microdroplet source for deposition purposes. Appl Surf Sci 65/66:1–12ADSView ArticleGoogle Scholar
- Bischoff I, Akhmadaliev C, Mair AWR, Mair GLR, Ganetsos T, Aidinis CJ (2004) Investigation of a tin liquid metal ion source. Appl Phys A 79:89–92ADSView ArticleGoogle Scholar
- Jaworek A (2007) Micro- and nanoparticle production by electrospraying. Powder Technol 176:18–35View ArticleGoogle Scholar
- Stubig M, Schafer G, Ho T-M, Srama R, Grun E (2001) Laboratory simulation improvements for hypervelocity micrometeorite impacts with a new dust particle source. Planet Space Sci 49:853–858ADSView ArticleGoogle Scholar
- Shu A, Collette A, Drake K, Gruen E, Horanyi M, Kempf S et al (2012) 3 MV hypervelocity dust accelerator at the Colorado Center for Lunar Dust and Atmospheric Studies. Rev Sci Instrum 83:075108ADSView ArticleGoogle Scholar